The Chroma A W and A W electronic load mainframes accept the user-installable A series load modules for easy system configuration. user can perform on line voltage measurement controlled instructions via RS- C or GPIB The Chroma W and W electronic load. File, CHROMA Series, , Mainframes X, Operation Share technical documentation including product manuals, getting started guides, and.
|Published (Last):||16 November 2015|
|PDF File Size:||9.6 Mb|
|ePub File Size:||20.57 Mb|
|Price:||Free* [*Free Regsitration Required]|
You can return the Load to local control from remote control by pressing LCL key. Chroma A Programmable Electronic Load integrates microprocessor capabilities into mabual load module and mainframe to provide simple and accurate parallel operation to optimize the speed and control among multiple load modules.
Special disclaimers apply to these instruments. The typical temperature coefficient is ppm. Overtemperature Each Load has an mmanual protection circuit, which will turn off the load if internal temperature exceeds safe limit.
Trademark Information is a registered trademark of. The ChromaW andW electronic load mainframes accept the user- installable series load modules for easy system configuration and fit 19″ instrument rack. It depends on the power rating of the module. Chroa Load module will display ocp when overcurrent protection occurs.
The Load module will display ovp when overvoltage protection occurs. Compatibility WitH series.
The built-in resistive load current sensing circuit is capable of measuring current with an accuracy of 0. The major considerations in making input connections are the wire size, length and polarity. DCS Series 1 kw, 1.
You may set the conducting voltage Von to solve the problems. Remote sensing compensates for voltage drop in applications that require long lead lengths. There are no upper and lower limits that would cause an error.
L for safety and airflow Channel Number Cjroma channel number of a specific Load is determined by the location of that module in relation to the farthest left side of Mainframe. Power Electronics Testings Test Solution www. Please refer to the System Installation Manual for information on limitations regarding product use and function and information on More information. If input voltage is majual the maximum of low range, you must select the high range.
Chroma 6314A 1200W Electronic Load Mainframe for 4 Load Modules
Products You May Like. Trademark Information is a registered trademark of More information. All fields are required. Determine the position for the Hands-free phone in the elevator phone More information.
Makes no warranty of any kind with regard to this manual, including, but not limited to, the implied warranties of merchantability and fitness for a particular purpose. The RH measures and displays Temperature. For example, the test of a battery charger can be simulated easily by setting the load to operate in constant voltage. Home About Us Contact Us.
All of the protection features will latch when they are tripped. Keep these More information.
Local operation on front chrmoa keypad. Do not attempt to disassemble the unit. Daker DK 1, 2, 3 kva. The overtemperature OT and temperature error TE status register bits are set when the OT condition occurs, and will remain set till they are reset.
The system is configured by plugging the user selectable load modules into the system mainframe.
Instrument Control (GPIB, Serial, VISA, IVI)
Here Load means the electronic load modules of Chroma series while Mainframe theelectronic load mainframes. This warranty is in lieu of all other warranties, expressed or implied, and no representative or person is authorized to represent or assume for Chroma chromma liability in connection with the sale of our products other than set forth herein.
kangas :: Chroma manual
If the input current of Load is over 40 Amps, you must use multiple terminals of connections. High voltage used in some instruments may be dangerous if misused. Operation Table of Contents 1. If the factory set switch on this label does not correspond to your nominal chrma voltage, turn the Mainframe power off, and disconnect the power cord.
Without the written permission of the copyright holder, More information. An error will occur when data are over the maximum or minimum value Constant Current Mode Figure Constant Current Mode In CC mode, the Load will sink a current in accordance with the programmed value regardless of input voltage.